Wide blade, axial flow pump

ABSTRACT

A blood pump comprises a pump housing; a rotor positioned in the housing and comprising an impeller having a hydrodynamic surface for pumping blood; and a motor including a plurality of magnets carried by the impeller, plus a rotor stator, including an electrically conductive coil located adjacent to or within the housing. The impeller comprises radially outwardly extending, bladelike projections that define generally longitudinally extending spaces between the projections. The shape of the projections and the spaces therebetween tend to drive blood in the spaces in an axial direction as the impeller is rotated. The spaces collectively have a total width along most of their lengths at the radial periphery of the rotor, that is substantially equal to or less than the collective width of the projections along most of their lengths at the radial periphery. Thus, the bladelike projections are thicker, achieving significant advantages.

BACKGROUND OF THE INVENTION

The known axial flow pumps for blood have the advantage of narrow width, when compared with radial flow pumps. Axial flow pumps typically have a cylindrical housing with an inlet at one end, an outlet at the opposite end, and a rotor within the housing which has impeller blades attached to the rotor. Thus, as the rotor rotates, the blades add work to the fluid, propelling the fluid through one end of the housing.

A suspension system is provided to maintain the rotor in desired position within the housing, and a motor is provided to spin the rotor. Blood flows between the blades, being propelled through the pump by hydrodynamic forces transferred by the blade surfaces.

The blood then leaves the pump, flowing parallel to the axis of rotation of the rotor. Typically in the prior art, the rotor is suspended by mechanical bearings or bushings, with a rotor shaft protruding through the pump housing to a motor drive mechanism. Magnetic suspension is also known, as in U.S. Pat. Nos. 6,368,083 and 5,840,070.

Typically, axial blood flow pumps have used a thin blade design, with the motor magnets being placed either in the rotor shaft far away from the surrounding stator as in pumps by Jarvik and Incor, or they use small magnets placed within the thin blades, as in the MicroMed pump. Both of these approaches tend to reduce the motor torque capacity and efficiency, and they use mechanical rotor support involving abutting surfaces that move relative to each other in rotation.

By this invention a new utilization of wide (thick), blade-like projections on a rotor in an axial flow configuration is provided for a blood pump, to provide a pump which is mechanically wearless, and can have improved torque. Blood pumps, whether internally or externally located, must exhibit low hemolysis, good resistance to thrombosis, adequate system efficiency, and very high reliability for the expected duration of use for the device. Internally located blood pumps are also subject to anatomical compability design constraints, and the need for elimination of mechanical wear and associated failure modes in order to provide successful, long-term, implantable devices. The pump of this invention can achieve the above. Also, the pump can be sealless.

While the pump of this invention is described in terms of a blood pump, it is also contemplated that the pump might be used for pumping chemically difficult fluids, where a sealless design is highly desirable, and the fluid must be gently handled for various reasons, for example because it is unstable to mechanical stress, causing decomposition and even explosiveness, or because it is another complex, biological fluid besides blood, having critical stability parameters, like blood.

DESCRIPTION OF THE INVENTION

In accordance with this invention a blood pump is provided which comprises a pump housing; and a rotor positioned in the housing and comprising an impeller having a hydrodynamic surface for pumping blood. A motor is provided, the motor having a plurality of magnetic poles of a magnet or magnets carried by the impeller. A motor stator is provided, which includes an electrically conductive coil located adjacent to or within the housing.

Hydrodynamic bearing surfaces may also be present, being symmetrically located around the impeller. The term “hydrodynamic bearing surfaces” implies that the bearing surface is acting against fluid to impart forces to the rotor, which helps to position the rotor.

The impeller comprises radially outwardly extending, blade-like projections that define generally longitudinally extending spaces between the projections. The projections are shaped to form curves in the spaces of a shape tending to drive blood in an axial direction as the impeller is rotated. By this invention, the spaces collectively have a total width (i.e., the entire sum of the widths of the spaces) that is substantially equal to or less than the collective, total widths of the projections themselves. This measurement is taken at the radial periphery of the rotor.

Thus, in accordance with this invention, the bladelike projections are each much wider, particularly at their peripheries, than in analogous prior art axial flow blood pumps, and the longitudinal spaces or channels between the projections are narrower particularly at the peripheries, than in the prior art. This permits the emplacement of larger motor magnets enclosed inside of the projections, to increase the magnetic flux. Also, the motor air gap can be reduced, when the motor magnets are near the outer tip of each projection, being closer to the motor stator. This increases magnet flux area, which, with the reduced air gap, improves the motor torque capacity and electromagnetic efficiency.

The wide, blade-like projections also preferably have hydrodynamic thrust bearings at radially outer edges of the bladelike projections, with the thrust bearings having sufficient surface area for rotor radial support. The hydrodynamic thrust bearings may work by providing a decreasing flow area in the direction of rotation, and are well known to the art generally, as in U.S. Pat. No. 5,840,070. The external work provided by the rotor forces blood flow through a decreased or constricted area created by the hydrodynamic thrust bearings. This results in increased fluid pressure upstream of the constriction, which pressure acts against the surface area and produces a net force for radial support. This hydrodynamic force that is thus created on the outer edges of the rotor projections can resist magnetic forces from the motor and any dynamic, radial shock loading forces.

Typically, the bladelike projections define longitudinally extending spaces between them, with sidewalls having transverse sections that mostly have generally parallel sides, as in FIG. 5.

It is also preferred for the bearings to each define a bearing surface with shrouds carried at ends of the bearing surface, typically at the radially outer face of each hydrodynamic bearing. These bearing shrouds can reduce the amount of end fluid leakage, and can allow the development of higher pressure levels. Fluid leakage can limit the amount of pressure that a hydrodynamic bearing can generate. The reduction of such end leakage to acceptable levels by means of the shrouds can almost double the load carrying capacity for the bearings.

Also, a pressure relief surface, which may be a diverging area downstream of a thrust bearing, can be added to reduce the level of hemolysis of the blood being pumped.

Additionally, the hydrodynamic thrust bearings which are located on the outer periphery of each rotating projection can also be provided with good fluid washing, since centrifugal forces tend to push fluid toward the outer periphery of the housing interior, providing increased blood flow, which can improve the pump's resistance to thrombosis. Hydrodynamic bearings which are closer to the axis of rotation will have reduced surface washing, resulting in a greater possibility of blood coagulation. Thus, since by this invention, conditions are provided that reduce blood coagulation, a lower amount of anticoagulant may be used with the blood pump and patient, which may result in fewer patient adverse side effects.

Also, at least one magnetic bearing system may be provided, as well as the hydrodynamic bearings, to help to position the rotor in its desired position within the housing. Magnetic bearing systems work by having two sets of magnets that repel each other. One set can be located outside or within the tubular housing, and the other, opposing magnets can be located within the wide, bladelike projections of the rotor.

The magnets mounted in the projections may be permanent magnets. The electric motor used may be of radial flux gap design, so that axial magnetic forces assist in holding the rotor in position.

The rotor in accordance with this invention does not necessarily require additional supporting structures upstream or downstream thereof, in the circumstance where the axial, magnetic forces and the thrust bearings are sufficient to maintain the rotor in desired position during operation.

The motor design may comprise a three phase, brushless DC motor, with the motor stator being positioned outside of the housing that carries the blood, which housing is axially aligned with the rotor. The stator contains the motor windings, and may have a back iron design which is consistent with a typical radial flux gap motor design. A large, permanent magnet may be carried because the projections are thick, to provide a strong electromagnetic coupling, and also it can provide the necessary axial stiffness to maintain the rotor in position.

If desired, the stator can comprise a separate, hermetically sealed motor that slides over a tubular housing into position, and is secured thereto. By this means, system efficiency can be improved, and any current loses can be reduced. Laser welding is one possibility for obtaining a hermetic seal if the stator is built into the housing.

The device controller can run the motor at a set rotational speed, which may be set, for example by the attending physician, or it may follow a physiological control algorithm. Pulse width modulation can also be used for speed control.

The permanent magnets at the periphery of the rotating projections may be covered by peripheral cover caps. These cover caps may also provide an added function by defining weight reduction open spaces such as holes, which may be formed to achieve balance of the rotor.

The housing may comprise a simple tube, with a rotor being slid into place and held there by magnetic attraction to the enveloping stator. The housing may be made of a biocompatible material such as titanium or ceramic. A braised weld ring to the housing outer surface may be used to secure the motor stator. However, the housing does not have to be a constant diameter tube. For example, it might carry a diverging section in the direction of flow, mated to a rotor having a tapered front section containing a hydrodynamic thrust bearing, for additional protection against axial shock loading. Such alignment of a housing diverging section and the motor design could provide a magnetic, axial preload to help ensure that the rotor maintains its position even if it is normally suspended by magnetic force created between the rotor and the motor stator within a non-magnetic housing.

Thus, a wide-bladed, axial flow pump optionally typically utilizing shrouded hydrodynamic thrust bearings is provided, having significant advantage in the pumping of blood. The motor may be integrated within the rotor's bladelike projections, allowing for a compact device with improved system efficiency. The hydrodynamic thrust bearings on the peripheral, blade-like projection surfaces serve to place the main, wearless suspension system component in a region of good washing for increased resistance to thrombosis. Alternative configurations can exclusively use magnetic bearings, since bearing magnets can easily be incorporated within the wide, blade-like projections, or it may be coupled with a sloped forward section, containing a thrust bearing for increased axial thrust resistance.

Typically with this design, the height of the blade-like projections is greater than in comparable and conventional thin blade designs, to make up for a loss of flow area in the circumferential direction. For example, for heart pump uses, the height of such blades from the axis of rotation to their outer faces may be at least 2 mm, up to typically about 10 mm.

The design of this invention is more tolerant of flow variations than previous, thin blade designs, since the flow of blood (or other fluid) can more easily adjust to off-design incident angles, in the absence of upstream flow straighteners, than do thinner blades, which tend to be highly tuned to a particular flow condition. Typically, 2-8 of the wider blades of this invention are provided to a single rotor.

Typically, the permanent motor magnets which are located within the wide, blade-like projections may be selected for magnetic properties, length, and cross-sectional area which provides good electromagnetic coupling with the stator. Because of the large dimensions of the blade-like projections, this particular design becomes easier to effect.

The preferred configurations are “sealless” since the rotor is driven by the motor stators separated from the rotor through a sealed, typically tubular housing.

The hydrodynamic thrust bearings near the leading and/or trailing edge portions of the rotor may be sloped to provide both radial and axial support, and may be useful to increase the device resistance to shock loading. A single, hydrodynamic thrust bearing can be used on the outer surface of each blade-like projection. Alternatively, separate bearings could be used in the leading edge region, the rotor mid-section, and/or the trailing edge region respectively. Typically, hydrodynamic thrust bearings can be placed near the leading edge and near the trailing edge, for rotor stability. The hydrodynamic bearing can be installed perpendicular to the axis of rotation. Also they may be flow aligned in a helix fashion, to improve surface washing as they operate. An optional pressure relief surface downstream of the bearing to reduce hemolysis may also be provided to each bearing if appropriate. This comprises a slightly diverging section to decrease the flow velocity in the direction of rotation, downstream of the hydrodynamic bearing.

Alternatively, magnetic bearings may be used to replace the hydrodynamic thrust bearings, to provide an all magnetic system, if desired. These magnetic bearings could be positioned either forward or aft of the motor magnets.

Unlike designs using thinner blades, upstream and downstream struts which serve as flow straighteners and diffusers are not typically required in devices in accordance with this invention. The absence of these upstream and downstream flow straighteners permits a simpler mechanical design, with fewer axial tolerance concerns associated with the placement of these flow straighteners or diffusers. Also, in the absence of the flow straighteners and diffusers, the device tolerance to off design conditions is increased, by allowing the flow to condition itself prior to entering or leaving the rotor. This reduces hemolysis and improves resistance to thrombosis for blood pump applications.

As previously stated, the pumps in accordance with this invention can also be used for other fluids, for example other biological fluids, or other critical fluids of chemical processes and the like.

Alternatively, the wide, blade-like projection-using pump of this invention may be utilized with a mechanical pivot bearing rotor suppression system for greater axial constraint, while taking advantage of the greater motor efficiency of the wider, blade-like projections in accordance with this invention.

Typically, when a single stage, axial blood pump in accordance with this invention is used, one specific embodiment would be a rotor having four blade-like projections placed within a continuous, straight housing tube, with the motor being located outside of the housing. The blade-like projections would be designed to have a hydrodynamic thrust bearing near the leading and trailing edges of each projection. The device could be free of upstream and downstream support structures. The motor could be a toroidal, three phase, and wye connected design, in one preferred embodiment. If such a device is being designed for a permanent heart ventricular assist device, it could be a cylindrical device having a 10 millimeters outer diameter and 20 millimeters length, providing flow rates of 2-10 liters per minute against physiologic, differential blood pressures.

Another housing configuration would use a sloped surface for the rotor leading edge, to provide both axial and radial support with the motor axial magnetic stiffness providing axial support. In another embodiment, a similar, sloped surface could be provided on the rotor trailing edge to provide both axial and radial support.

Alternately, a split housing configuration might be provided, with sloped surfaces at both the rotor and leading and trailing edges, to provide axial and radial support. Also, the inflow design of the tubular housing could have a converging section for the hydrodynamic thrust bearings to run against, to provide axial and radial support.

The blood pump of this invention might be placed in line with a cannulation system, and with the pump being located within the chest cavity of a patient, such as the pericardial space, abdomen, or subcutaneously near the skin, in a manner similar to pacemaker implantation. Likewise, the pump may be kept external to the body for shorter term vascular circulatory support. Also two single stage pumps in accordance with this invention could be used in tandem to provide bi-ventricular support, or even total circulation for the patient in the manner of a full, artificial heart.

Where a smaller diameter blood pump is desired, this can be achieved by the use of multiple blood pump stages connected in series. If desired, stator blades may be provided between the various blood pump stages to de-swirl the flow so that more hydraulic work can be added to the fluid. The stator blades may be of any desired number and typically of a traditional thin blade design. However, if desired, stator blades may carry hydrodynamic thrust bearings, and may be used to provide axial support to the rotor. Also, stator blades may be of wide configuration and may include permanent magnets as an integral portion of a magnetic bearing, with other magnets that work with the magnets of the stator blades being located within a rotor shaft or in a rotor blade-like projection.

Also, in a multiple stage pump, single or multiple hydrodynamic thrust bearings may be located on each pump unit of the multiple pump system, or hydrodynamic thrust bearings may be absent from some of the stages if the additional radial support is not required, which of course would increase device efficiency.

Axial alignment of the motor and hydraulic pump stages in a multiple pump may be the same, or leading and trailing motor stages may be located inboard or outboard of the leading and trailing stages, to provide extra axial magnetic support.

As previously stated, upstream and downstream flow straighteners and diffusers are generally not necessary, even in multiple stage units, especially for congealing fluids, such as blood, but they could be included for additional gains and device efficiency or additional axial constraint.

For example, a typical multiple stage pump unit having rotors with wide, blade-like projections may comprise an axial flow blood pump which would have two stages for example, each with a rotor having four of the wide projections, placed within a cylindrical housing, with the motor also having stages and located on the outside of the housing in line with the rotor stages. The wide blades would typically have a hydrodynamic thrust bearing near the leading and trailing edges of the blade. No upstream or downstream support structures would generally be necessary. The motor could be a toroidal, three phase, and wye connected design.

A size for a permanent ventricular assist device of multistage configuration as described above could be an outer diameter of six millimeters and a length of 15 millimeters, to provide flow rates of 2-8 liters per minute against physiological differential pressures, as previously described. Also, such a device could be used for a peripheral vessel blood insertion pump, operating outside of the body. Also, a multiple stage pump, as previously stated, could provide bi-ventricular support and even total artificial heart action.

The pump utilizes hydrodynamic thrust bearings for radial support and the magnetic bearings for axial support, primarily resulting in a system which has no mechanical wear, since no rubbing of solid surfaces takes place in a significant manner.

DESCRIPTION OF THE DRAWINGS

In the drawings, FIG. 1 is an enlarged, longitudinal sectional view of an implantable, sealless, rotary blood pump in accordance with this invention.

FIG. 2 is a further, enlarged elevational view of the rotor of the pump rotor of FIG. 1.

FIGS. 3 and 4 are additional side views of the rotor of FIG. 2 in differing positions.

FIG. 5 is a sectional view taken along line 5-5 of FIG. 2, with internal parts omitted.

FIG. 6 is a perspective view of an alternative embodiment of a rotor usable in the pump of this invention.

FIG. 7 is an enlarged, rear perspective view of another rotor which is similar to the rotor of FIG. 2.

FIG. 8 is a front perspective view of the rotor of FIG. 7.

FIG. 8A is an enlarged, fragmentary, perspective view of a portion of the rotor of FIG. 7

FIG. 9 is a plan view, taken partially in longitudinal section, showing a ganged series of blood pumps, each of the design in accordance with this invention to provide a pump with greater power.

FIG. 10 is an enlarged, longitudinal sectional view of a revised embodiment, similar to FIG. 1.

DESCRIPTION OF SPECIFIC EMBODIMENTS

Referring to FIGS. 1-5, a blood pump 10 is disclosed, comprising a generally cylindrical pump housing 12, which may be made of a non-thrombogenic, rigid, strong material such as titanium and/or a suitable ceramic material. Rotor 14 is positioned within the lumen of pump housing 12, and acts as an impeller, having a hydrodynamic surface, specifically a series of hydrodynamic surfaces 16 that tend to propel blood in an axial direction (as indicated by arrow 18) as rotor 14 is rotated clockwise. Blood pump 10 may be connected to the patient's vascular system.

Rotor/impeller 14 comprises radially outwardly extending, blade-like projections 20 having side walls 16 that define generally longitudinally extending spaces 22 between the projections 20. The projections 20 and their side walls 16 are shaped to form curves in the longitudinally extending spaces 22 which are of a shape tending to drive blood in axial direction 18 as rotor/impeller 14 is rotated (clockwise in the embodiment of FIG. 1).

It will be noted, particularly from FIG. 5, that the longitudinally extending spaces 22 collectively have, adjacent to radially outer periphery 23 at the outer circumference of rotor 14, a collective, total circumferential width that is substantially less than the collective, total circumferential width of the projections 20 at the same radially outer periphery 23. This is illustrated by peripheral width 26, illustrated on one of the longitudinally extending spaces 22 in FIG. 5, when compared with peripheral width 28 of adjacent, blade-like projections 20. Collectively, the four widths 26 of each of the spaces 22 comprise a collective, total width of all four longitudinally extending spaces 22. Four times the distance of arc 28 represents the collective, total width of the four blade-like projections 20. It can be readily seen that the collective total width of the longitudinally extending spaces 22 is substantially less at periphery 23 than the collective, total width of the respective blade-like projections 20, in the embodiment of FIGS. 1-5.

It is preferred for transverse sections (FIG. 5) of longitudinally extending spaces 22 to have generally parallel side walls 16, although it can also be seen from FIG. 1 and other drawings that the overall width of longitudinally extending spaces 22 may vary along their lengths, being particularly somewhat narrower at upstream areas 30, and wider at downstream areas 32, as shown in FIG. 1.

Thus, it can be seen from particularly FIG. 1 that clockwise rotation of rotor 14 will result in a flow of blood within the lumen of housing 12 from left to right in direction 18.

Blood pump 10 further comprises a motor, which includes a plurality of relatively large motor magnets 34 (FIG. 2) respectively carried in the thick, wing-like projections 20 under magnet cover 35 (FIG. 2). The motor also comprises a motor stator 36 (FIG. 1), including an electrically conductive coil 38, within an enclosure 40, which surrounds housing 12 and rotor 14, and serves to rotate rotor 14 by the conventional application of electric power to coil 38, which is converted via magnetic force to torque, causing rotor 14 to rotate clockwise. The specific technology for accomplishing this may be similar to that which is well known in the prior art.

FIGS. 1, 2, 7, and 8 show a radially outer face 42 of a blade-like projection 20, also showing a pair of hydrodynamic bearings 44, 46, which may be carried on magnet cover 35 in the embodiment of FIGS. 1-5, and which use fluid pressure to cause rotor 14 to be centered in the lumen of tubular housing 12 as rotor 14 rotates, in a manner generally shown in FIG. 1, without the need for physical bearings utilizing rubbing, solid surfaces. To facilitate the proper positioning, particularly longitudinal positioning, of rotor 14 within housing 12, a second set of magnets 48, 50 is shown. First magnets 48 (FIG. 2) are mounted in projections 20, under magnet cover 35, adjacent the forward end thereof in this embodiment, with their north poles facing outwardly. Second magnets 50 are carried on tubular housing 12 with their north poles facing inwardly, so that magnetic repulsion takes place between magnets 48 and magnets 50. Of course, the south poles could be directed to face each other in similar manner, to achieve a generally similar effect. Alternatively, magnets 50 may comprise a single, ring magnet or an electromagnetic coil.

Thus, rotor 14 rotates, being held away from the inner wall of housing 12 by hydrodynamic bearings 44, 46 on each of the wing-like projections 20. Longitudinal movement to the right, as in FIG. 1, of rotor 14 is restricted by the action of magnets 48, 50. At the rear of rotor 14, an inner, annular ring 52 is seen to project a bit inwardly from the inner wall cylinder housing 12, to limit the leftward motion of rotor 14. Projection 52 may, if desired, comprise an annular series of spaced projections, or it may comprise a solid ring, with hydrodynamic bearing 44 (FIGS. 2 and 7) serving to prevent contact between rotor 14 and ring 52 as the pump is operating with clockwise rotation of rotor 14. If desired, a similar annular ring 53 may be placed near the other end of housing 12 for similar purpose.

Cover 35 not only carries thrust bearings 44, 46, but it encloses and retains magnets 34, 48.

Of course, it is within the scope of this invention to design a rotor which can rotate in the counterclockwise direction, making use of the principles and advantages of this invention.

If desired, the stator 36 may comprise a separate, hermetically sealed, coil motor that slides over tubular housing 12 in position, and is secured thereto. Otherwise, stator and coil 38 may be integrally attached to housing 12.

Referring to FIGS. 7, 8, and 8A, details concerning hydrodynamic thrust bearings 44, 46 of rotor 14 a are shown in this embodiment, which is similar to the FIGS. 1-5 embodiment except as otherwise stated.

Each of thrust bearings 44, 46 define a recessed, curved outer surface 47 which forms a recessed end portion 49 relative to the outer face 42 of each projection 20, located at the forward end of each bearing 44, 46 from the viewpoint of the (clockwise) spin of the rotor 14 a, so that recessed end 49 forms a leading edge of rotation. The recessed surface 47 then tapers in a gradual, curved manner outwardly to the rear end 51 of each thrust bearing 44, 46, at which point, the bearing surface 47 is not recessed, or only very slightly recessed compared with end 49.

Thus, as the rotor rotates, the respective thrust bearings, 44, 46 on each projection 20 scoop blood into a cross-sectional recessed area that decreases going from end 49 to end 51, the effect of this being to pressurize the blood, and to thus repel each projection 20 from the inner wall of housing 12 as the rotor rotates. Since the rotor is spaced from the walls of housing 12, the pressurized blood is released by passing across end 51 and out the sides. Shroud walls 53 (FIG. 8A) connect recessed surface 47 with the rest of projection outer face surface 42.

Bearing 44 operates in a manner similar to bearing 46. In the embodiment of FIGS. 1-5 the bearings 44, 46 also operate in a similar manner Pressure relief zone 55 is provided at the trailing rotary end of each rotating projection 20, to provide pressure relief.

Rotor 14 a of FIGS. 7-8A may be used as a substitute for rotor 14. However, rotor 14 a may be produced by either machining, molding, or casting a single piece of isotropic, ferromagnetic material, such as compression bonded neodymium or Alnico (aluminum-nickel alloy). After fabrication, the rotor may be treated with a conformal, protective polymer coating of an organic polymer such as Parylene, or silicone, to prevent against oxidation by forming a hermetic seal around the rotor. On top of this, a hard, lubricious protective coating may be applied over the conformal polymer coating, to protect against wear and abrasion. Such coatings may include chromium nitride, titanium-nitride, or other commercially available coatings such as ME92, Med Co 2000, or DLC.

Also, before or after the coating application, the rotor projections 20 may be alternatively magnetized N-S-N-S to form a salient pole rotor magnet so that, contrary to FIG. 1, separate magnets 48 and an outer cover are not present, and each entire rotor projection 20 is appropriately magnetized to operate in the motor of FIG. 1.

Referring to FIG. 6, another embodiment of a rotor 14 b for the blood pump of this invention is disclosed. Rotor 14 b is shown to have six blade-like projections 20 b, which are generally similar in structure and function to the blade-like projections 20 of the previous embodiments.

Referring to FIG. 9, a ganged series of blood pumps 60 is shown, having a common, cylindrical housing 62 for a series of rotors 14 c, each carried on a common shaft 64 so that the rotors rotate as one along with the shaft 64. Each of the rotors 14 c have radially outwardly extending, blade-like projections 20 c, which may be of a general design and shape similar to projections 20 of the previous embodiments, apart from modifications that result from their mounting on a common shaft 64, so that the rotors 14 c and their respective projections 20 c operate in a manner similar to rotor 14 and projections 20 of the previous embodiments. By this means, added pumping power can be provided in the form of a multiple stage pump, with the rotors in series connection. Accordingly, a high capacity pump of smaller diameter can be provided.

Motor stators 36 c, comprising an electrically conductive coil as in the previous embodiments, are provided, one for each rotor, so that the respective rotors perform in a manner similar to that of the previous embodiments, but for their connection with the common shaft. The rotors 14 c and stators 36 c may be of the same design as any of the previous embodiments, but for the changes specifically mentioned here.

Stator blades 66 are respectively mounted on the inner wall of pump housing 62, extending inwardly therefrom to de-swirl the flow, permitting more hydraulic work to be added to the fluid. Any desired number of these generally radially extending blades 66 may be provided.

Referring to FIG. 10, it is a drawing of a revised embodiment of the pump of FIG. 1, similar in substantially all respects except for the addition of retention sleeve 70, which provides longitudinal retention of rotor 14 in a mechanical way so that magnets 48, 50 are unnecessary, although they may also be present. Also, the flow outlet defined by sleeve 70 is of course of reduced diameter from the embodiment of FIG. 1, rendering this arrangement suitable as a pediatric version of the pump of this invention.

The above has been offered for illustrative purposes only, and is not intended to limit the scope of the invention of this application, which is as defined in the claims below. 

1. A blood pump, comprising: a pump housing; a rotor positioned in said housing and comprising an impeller having a hydrodynamic surface for pumping blood; a motor including a plurality of magnetic poles carried by the impeller, and a rotor stator including an electrically conductive coil located adjacent to or within said housing; and said rotor comprising radially outwardly extending, bladelike projections that define generally longitudinally extending spaces between said projections, said projections being shaped to form curves in said spaces of a shape tending to drive blood in said spaces in an axial direction as the impeller is rotated, said spaces collectively having a total width along most of their lengths, at the radial periphery of the rotor, that is substantially equal to or less than the collective total width of said projections along most of their lengths at said radial periphery.
 2. The blood pump of claim 1 in which radially outer faces of said projections carry hydrodynamic thrust bearings.
 3. The blood pump of claim 2 in which said bearings each define a bearing surface with shroud walls at sides of said bearing surface transverse to the rotation axis of the rotor.
 4. The blood pump of claim 2 in which at least one magnetic bearing system is also provided to help to position said rotor.
 5. The blood pump of claim 4 in which said at least one magnet comprises permanent magnets mounted in said bladelike projections, the motor being of the radial flux gap design, whereby axial, magnetic forces hold the rotor in position.
 6. The blood pump of claim 5 in which said permanent magnets are covered with peripheral cover caps.
 7. The blood pump of claim 6 in which said peripheral cover caps define weight reducing open spaces.
 8. The blood pump of claim 1 in which the motor comprises a separate, hermetically sealed motor that slides over a tubular housing into position, and is secured thereto.
 9. The blood pump of claim 1 in which said housing is cylindrical.
 10. The blood pump of claim 1 in which said at least one magnet comprises permanent magnets mounted in said projections.
 11. The blood pump of claim 1 in which said housing is made from biocompatible titanium or ceramic.
 12. The blood pump of claim 1 which comprises a plurality of said rotors positioned within said pump housing, and each engaged by one of a plurality of said motors, in series flow, for increased pumping pressure capacity.
 13. The blood pump of claim 1 in which said projections define said longitudinally extending spaces with sidewalls having transverse sections that mostly have generally parallel sides.
 14. A blood pump comprising: a pump housing; a rotor positioned at said housing and comprising an impeller having a hydrodynamic surface for pumping blood; a motor including a plurality of magnets carried by the impeller and a stator including an electrically conductive coil located adjacent to or within said housing; and said rotor comprising radially outwardly extending, blade like projections that define generally longitudinally extending spaces between said projections, said projections being shaped to form curves in said spaces of a shape tending to drive blood in said spaces in an axial direction as the impeller is rotated.
 15. The blood pump of claim 14 in which said projections have radially outer faces which carry hydrodynamic thrust bearings, and in which at least most of the volume of each of said projections is magnetized to permit rotation of said rotor by the motor.
 16. The blood pump of claim 14 in which the motor comprises a separate, hermetically sealed motor that slides over said pump housing into position, and is secured thereto, said housing being cylindrical.
 17. The blood pump of claim 16 in which said housing is made from biocompatible titanium alloy or ceramic.
 18. The blood pump of claim 14 which comprises a plurality of said rotors positioned within said pump housing, and each being engaged by one of a plurality of said motors, in series flow, for increasing pumping pressure capacity.
 19. The blood pump of claim 18 in which radially extending stator plates are provided between said motors and rotors.
 20. The blood pump of claim 14 in which said projections defining said longitudinally extending spaces have side walls with transverse sections that mostly have generally parallel sides.
 21. The blood pump of claim 14 in which said rotor carries a conformal polymer coating to prevent oxidation by forming a hermetic seal around said rotor.
 22. The blood pump of claim 21 which carries a hard, lubricious protective coating over said conformal polymer coating to protect from wear and abrasion.
 23. The blood pump of claim 14 which carries a hard, lubricious protective coating over said rotor to protect said rotor from wear and abrasion.
 24. The blood pump of claim 14 in which said longitudinally extending spaces collectively have a total width at the radial periphery of the rotor that is substantially equal to or less than the collective total width of said projections at said radial periphery. 